This site displays a prototype of a “Web 2.0” version of the daily Federal Register. It is not an official legal edition of the Federal Register, and does not replace the official print version or the official electronic version on GPO’s govinfo.gov.
The documents posted on this site are XML renditions of published Federal Register documents. Each document posted on the site includes a link to the corresponding official PDF file on govinfo.gov. This prototype edition of the daily Federal Register on FederalRegister.gov will remain an unofficial informational resource until the Administrative Committee of the Federal Register (ACFR) issues a regulation granting it official legal status. For complete information about, and access to, our official publications and services, go to About the Federal Register on NARA's archives.gov.
The OFR/GPO partnership is committed to presenting accurate and reliable regulatory information on FederalRegister.gov with the objective of establishing the XML-based Federal Register as an ACFR-sanctioned publication in the future. While every effort has been made to ensure that the material on FederalRegister.gov is accurately displayed, consistent with the official SGML-based PDF version on govinfo.gov, those relying on it for legal research should verify their results against an official edition of the Federal Register. Until the ACFR grants it official status, the XML rendition of the daily Federal Register on FederalRegister.gov does not provide legal notice to the public or judicial notice to the courts.
Food and Drug Administration, HHS.
The Food and Drug Administration (FDA) is announcing the availability of the guidance entitled “Class II Special Controls Guidance Document: Low Energy Ultrasound Wound Cleaner.” This guidance document has been developed as a special control guidance document to support the classification of the low energy ultrasound wound cleaner into class II (special controls). The device is intended for the cleaning and maintenance debridement of wounds. This guidance document describes a means by which the low energy ultrasound wound cleaner may comply with the requirement of special controls for class II devices. Elsewhere in this issue of the Federal Register, FDA is publishing a final rule to classify the low energy ultrasound wound cleaner into class II (special controls). The guidance document is immediately in effect as the special control for the low energy ultrasound wound cleaner, but it remains subject to comment in accordance with the agency's good guidance practices (GGPs).
Submit written or electronic comments on this guidance at any time. General comments on agency guidance documents are welcome at any time.
Submit written requests for single copies on a 3.5″ diskette of the guidance document entitled “Class II Special Controls Guidance Document: Low Energy Ultrasound Wound Cleaner” to the Division of Small Manufacturers, International, and Consumer Assistance (HFZ-220), Center for Devices and Radiological Health, Food and Drug Administration, 1350 Piccard Dr., Rockville, MD 20850. Send one self-addressed adhesive label to assist that office in processing your request, or fax your request to 301-443-8818. See the SUPPLEMENTARY INFORMATION section for information on electronic access to the guidance.
Submit written comments concerning this guidance to the Division of Dockets Management (HFA-305), Food and Drug Administration, 5630 Fishers Lane, rm.1061, Rockville, MD 20852. Submit electronic comments to http://www.fda.gov/dockets/ecomments. Identify comments with the docket number found in brackets in the heading of this document.Start Further Info
FOR FURTHER INFORMATION CONTACT:
David B. Berkowitz, Center for Devices and Radiological Health (HFZ-410), Food and Drug Administration, 9200 Corporate Blvd., Rockville, MD 20850, 301-594-3090, ext. 152.End Further Info End Preamble Start Supplemental Information
The guidance document “Class II Special Controls Guidance Document: Low Energy Ultrasound Wound Cleaner” has been developed as a Start Printed Page 67490special control guidance document to support the classification of the low energy ultrasound wound cleaner into class II (special controls). This device is intended for the cleaning and maintenance debridement of wounds. On April 29, 2004, Celleration, Inc., submitted a petition requesting classification of the Celleration MIST Therapy SystemTM under section 513(f)(2) of the Federal Food, Drug, and Cosmetic Act (the act) (21 U.S.C. 360c(f)(2)).
Elsewhere in this issue of the Federal Register, FDA is publishing a final rule classifying the low energy ultrasound wound cleaner into class II (special controls) under section 513(f)(2) of the act. This guidance document will serve as the special control for the low energy ultrasound wound cleaner device. Section 513(f)(2) of the act provides that any person who submits a premarket notification under section 510(k) of the act (21 U.S.C. 360(k)) for a device that has not previously been classified may, within 30 days after receiving an order classifying the device in class III under section 513(f)(1) of the act, request FDA to classify the device under criteria set forth in section 513(a)(1) of the act. FDA shall, within 60 days of receiving such a request, classify the device by written order. This classification shall be the initial classification of the device. Within 30 days after the issuance of an order classifying the device, FDA must publish a notice in the Federal Register announcing such classification. Because of the timeframes established by section 513(f)(2) of the act, FDA has determined, under § 10.115(g)(2) (21 CFR 10.115(g)(2)), that is not feasible to allow for public participation before issuing this guidance as a final guidance document. Therefore, FDA is issuing this guidance document as a level 1 guidance document that is immediately in effect. FDA will consider any comments that are received in response to this notice to determine whether to amend the guidance document.
II. Significance of Guidance
This guidance is being issued consistent with FDA's GGPs regulation in § 10.115. The guidance represents the agency's current thinking on the low energy ultrasound wound cleaner for the cleaning and maintenance debridement of wounds. It does not create or confer any rights for or on any person and does not operate to bind FDA or the public. An alternative approach may be used if such approach satisfies the requirements of the applicable statute and regulations.
III. Electronic Access
To receive “Class II Special Controls Guidance Document: Low Energy Ultrasound Wound Cleaner” by fax, call the CDRH Facts-On-Demand system at 800-899-0381 or 301-827-0111 from a touch-tone telephone. Press 1 to enter the system. At the second voice prompt, press 1 to order a document. Enter the document number 1302 followed by the pound sign (#). Follow the remaining voice prompts to complete your request.
Persons interested in obtaining a copy of the guidance may also do so by using the Internet. CDRH maintains an entry on the Internet for easy access to information including text, graphics, and files that may be downloaded to a personal computer with Internet access. Updated on a regular basis, the CDRH home page includes device safety alerts, Federal Register reprints, information on premarket submissions (including lists of approved applications and manufacturers' addresses), small manufacturer's assistance, information on video conferencing and electronic submissions, Mammography Matters, and other device-oriented information. The CDRH Web site may be accessed at http://www.fda.gov/cdrh. A search capability for all CDRH guidance documents is available at http://www.fda.gov/cdrh/guidance.html. Guidance documents are also available on the Division of Dockets Management Internet site at http://www.fda.gov/ohrms/dockets.
IV. Paperwork Reduction Act of 1995
This guidance contains information collection provisions that are subject to review by the Office of Management and Budget (OMB) under the Paperwork Reduction Act of 1995 (the PRA) (44 USC 3501-3520). The collections of information addressed in the guidance document have been approved by OMB in accordance with PRA under the regulations governing premarket notification submissions (21 CFR part 807, subpart E, OMB control number 0910-0120). The labeling provisions addressed in the guidance have been approved by OMB under OMB control number 0910-0485.
Interested persons may submit to the Division of Dockets Management (see ADDRESSES) written or electronic comments regarding this document. Submit a single copy of electronic comments or two paper copies of any mailed comments, except that individuals may submit one paper copy. Comments are to be identified with the docket number found in brackets in the heading of this document. Received comments may be seen in the Division of Dockets Management between 9 a.m. and 4 p.m., Monday through Friday.Start Signature
Dated: October 5, 2005.
Linda S. Kahan,
Deputy Director, Center for Devices and Radiological Health.
[FR Doc. 05-22069 Filed 11-4-05; 8:45 am]
BILLING CODE 4160-01-S