Food and Drug Administration, HHS.
The Food and Drug Administration (FDA) is classifying the Immunomagnetic Circulating Cancer Cell Selection and Enumeration System device into class II (special controls). The special control that will apply to the device is the guidance document entitled “Class II Special Controls Guidance Document: Immunomagnetic Circulating Cancer Cell Selection and Enumeration System.” The agency is taking this action in response to a petition submitted under the Federal Food, Drug, and Cosmetic Act (the act) as amended by the Medical Device Amendments of 1976 (the amendments), the Safe Medical Devices Act of 1990 (the SMDA), the Food and Drug Administration Modernization Act of 1997 (FDAMA), and the Medical Device User Fee and Modernization Act of 2002 (MDUFMA). The agency is classifying the device into class II (special controls) in order to provide a reasonable assurance of safety and effectiveness of the device. Elsewhere in this issue of Start Printed Page 26037the Federal Register, FDA is publishing a notice of availability of a guidance document that is the special control for this device.
This rule is effective June 10, 2004. The classification was effective January 21, 2004.Start Further Info
FOR FURTHER INFORMATION CONTACT:
Nina Chace, Center for Devices and Radiological Health (HFZ-440), Food and Drug Administration, 2098 Gaither Rd., Rockville, MD 20850, 301-594-1293, ext. 138.End Further Info End Preamble Start Supplemental Information
In accordance with section 513(f)(1) of the act (21 U.S.C. 360c(f)(1)), devices that were not in commercial distribution before May 28, 1976, the date of enactment of the amendments, generally referred to as postamendments devices, are classified automatically by statute into class III without any FDA rulemaking process. These devices remain in class III and require premarket approval, unless and until the device is classified or reclassified into class I or II or FDA issues an order finding the device to be substantially equivalent, in accordance with section 513(i) of the act, to a predicate device that does not require premarket approval. The agency determines whether new devices are substantially equivalent to previously marketed devices by means of premarket notification procedures in section 510(k) of the act (21 U.S.C. 360(k)) and 21 CFR part 807 of FDA's regulations.
Section 513(f)(2) of the act provides that any person who submits a premarket notification under section 510(k) of the act for a device that has not previously been classified may, within 30 days after receiving an order classifying the device in class III under section 513(f)(1) of the act, request FDA to classify the device under the criteria set forth in section 513(a)(1) of the act. FDA shall, within 60 days of receiving such a request, classify the device by written order. This classification shall be the initial classification of the device. Within 30 days after the issuance of an order classifying the device, FDA must publish a notice in the Federal Register announcing such classification (513(f)(2) of the act).
In accordance with section 513(f)(1) of the act, FDA issued a notice on December 24, 2003, classifying the CellSearch Epithelial Cell Kit/Cell Spotter Analyzer in class III, because it was not substantially equivalent to a device that was introduced or delivered for introduction into interstate commerce for commercial distribution before May 28, 1976, or a device which was subsequently reclassified into class I or class II. On December 24, 2003, Veridex, LLC, submitted a petition requesting classification of the CellSearch Epithelial Cell Kit/Cell Spotter Analyzer under section 513(f)(2) of the act. The manufacturer recommended that the device be classified into class II (Ref. 1).
In accordance with 513(f)(2) of the act, FDA reviewed the petition in order to classify the device under the criteria for classification set forth in 513(a)(1) of the act. Devices are to be classified into class II if general controls, by themselves, are insufficient to provide reasonable assurance of safety and effectiveness, but there is sufficient information to establish special controls to provide reasonable assurance of the safety and effectiveness of the device for its intended use. After review of the information submitted in the petition, FDA determined that the CellSearch Epithelial Cell Kit/Cell Spotter Analyzer can be classified in class II with the establishment of special controls. FDA believes these special controls, in addition to general controls, will provide reasonable assurance of safety and effectiveness of the device.
The device is assigned the generic name immunomagnetic circulating cancer cell selection and enumeration system and is identified as a device consisting of biological probes, fluorochromes, and other reagents; preservation and preparation devices; and a semiautomated analytical instrument to select and count circulating cancer cells in a prepared sample of whole blood. This device is intended for adjunctive use in monitoring or predicting cancer disease progression, response to therapy, and for the detection of recurrent disease.
FDA has identified no direct risks to health when tests are used as an aid to monitoring and predicting cancer disease progression and response to therapy. However, failure of the test to perform as indicated, or an error in interpretation of results, could lead to misdiagnosis and improper treatment, improper patient management, improper treatment selection and dosing, and failure to identify circulating cancer cells. Consequently, FDA has identified the following risks to health associated with this type of device: (1) False negative, false low cancer cell count; and (2) false positive, false high cancer cell count. Therefore, in addition to the general controls of the act, the device is subject to special controls, identified as the guidance document entitled “Class II Special Controls Guidance Document: Immunomagnetic Circulating Cancer Cell Selection and Enumeration System.”
The class II special controls guidance document provides information on how to meet premarket (510(k)) submission requirements for the device including recommendations on validation of performance characteristics, including software validation; control methods; reproducibility; and clinical studies. FDA believes that following the class II special controls guidance document addresses the risks to health identified in the previous paragraph. Therefore, on January 21, 2004, FDA issued an order to the petitioner classifying the device into class II. FDA is codifying this classification by adding 21 CFR 866.6020.
Following the effective date of this final classification rule, any firm submitting a 510(k) premarket notification for an immunomagnetic circulating cancer cell selection and enumeration system will need to address the issues covered in the special controls guidance. However, the firm need only show that its device meets the recommendations of the guidance or in some other way provides equivalent assurance of safety and effectiveness.
Section 510(m) of the act provides that FDA may exempt a class II device from the premarket notification requirements under section 510(k) of the act, if FDA determines that premarket notification is not necessary to provide reasonable assurance of the safety and effectiveness of the device. For this type of device, FDA has determined that premarket notification is necessary to provide reasonable assurance of safety and effectiveness; therefore, the device is not exempt from premarket notification requirements. The device is used as an adjunct in monitoring or predicting cancer disease progression and response to therapy and for detection of recurrent disease. FDA's review of the test's sensitivity, specificity, and reproducibility with regard to key performance characteristics, test methodology and other relevant performance data, will ensure that acceptable levels of performance for both safety and effectiveness will be addressed before marketing clearance. Thus, persons who intend to market this type of device must submit to FDA a premarket notification, prior to marketing the device, which contains information about the immunomagnetic circulating cancer cell selection and enumeration system they intend to market.Start Printed Page 26038
II. Environmental Impact
The agency has determined under 21 CFR 25.34(b) that this action is of a type that does not individually or cumulatively have a significant effect on the human environment. Therefore, neither an environmental assessment nor an environmental impact statement is required.
III. Analysis of Impacts
FDA has examined the impacts of the final rule under Executive Order 12866 and the Regulatory Flexibility Act (5 U.S.C. 601-612), and the Unfunded Mandates Reform Act of 1995 (Public Law 104-4). Executive Order 12866 directs agencies to assess all costs and benefits of available regulatory alternatives and, when regulation is necessary, to select regulatory approaches that maximize net benefits (including potential economic, environmental, public health and safety, and other advantages; distributive impacts; and equity). The agency believes that this final rule is consistent with the regulatory philosophy and principles identified in the Executive order. In addition, the final rule is not a significant regulatory action as defined by the Executive order and so it is not subject to review under the Executive order.
The Regulatory Flexibility Act requires agencies to analyze regulatory options that would minimize any significant impact of a rule on small entities. Classification of these devices into class II will relieve manufacturers of the device of the cost of complying with the premarket approval requirements of section 515 of the act (21 U.S.C. 360e), and may permit small potential competitors to enter the marketplace by lowering their costs. The agency, therefore, certifies that the final rule will not have a significant impact on a substantial number of small entities. In addition, this final rule will not impose costs of $100 million or more on either the private sector or State, local, and tribal governments in the aggregate and, therefore, a summary statement of analysis under section 202(a) of the Unfunded Mandates Reform Act is not required.
FDA has analyzed this final rule in accordance with the principles set forth in Executive Order 13132. FDA has determined that the rule does not contain policies that have substantial direct effects on the States, on the relationship between the National Government and the States, or on the distribution of power and responsibilities among the various levels of government. Accordingly, the agency has concluded that the rule does not contain policies that have federalism implications as defined in the Executive order and, consequently, a federalism summary impact statement is not required.
V. Paperwork Reduction Act of 1995
This final rule contains no collections of information. Therefore, clearance by the Office of Management and Budget under the Paperwork Reduction Act of 1995 is not required.
The following reference has been placed on display in the Division of Dockets Management (HFA-305), Food and Drug Administration, 5630 Fishers Lane, rm. 1061, Rockville, MD 20852, and may be seen by interested persons between 9 a.m. and 4 p.m., Monday through Friday.
1. Petition from Veridex, LLC, dated December 24, 2003.Start List of Subjects
List of Subjects in 21 CFR Part 866End List of Subjects Start Amendment Part
Therefore, under the Federal Food, Drug, and Cosmetic Act and under authority delegated to the Commissioner of Food and Drugs,End Amendment Part Start Part
PART 866—IMMUNOLOGY AND MICROBIOLOGY DEVICESEnd Part Start Amendment Part
1. The authority citation forEnd Amendment Part Start Amendment Part
2. Section 866.6020 is added to subpart G to read as follows:End Amendment Part
(a) Identification. An immunomagnetic circulating cancer cell selection and enumeration system is a device that consists of biological probes, fluorochromes, and other reagents; preservation and preparation devices; and a semiautomated analytical instrument to select and count circulating cancer cells in a prepared sample of whole blood. This device is intended for adjunctive use in monitoring or predicting cancer disease progression, response to therapy, and for the detection of recurrent disease.
(b) Classification. Class II (special controls). The special control for this device is FDA's guidance document entitled “Class II Special Controls Guidance Document: Immunomagnetic Circulating Cancer Cell Selection and Enumeration System.” See § 866.1(e) for availability of this guidance document.Start Signature
Dated: April 26, 2004.
Linda S. Kahan,
Deputy Director, Center for Devices and Radiological Health.
[FR Doc. 04-10592 Filed 5-10-04; 8:45 am]
BILLING CODE 4160-01-S