Food and Drug Administration, HHS.
The Food and Drug Administration (FDA) is announcing the availability of the guidance entitled “Class II Special Controls Guidance Document: Low Level Laser System for Aesthetic Use.” This guidance document describes a means by which low level laser systems for aesthetic use may comply with the requirement of special controls for class II devices. Elsewhere in this issue of the Federal Register, FDA is publishing a final rule to classify low level laser systems for aesthetic use into class II (special controls). This guidance document is being immediately implemented as the special control for low level laser systems for aesthetic use, but it remains subject to comment in accordance with the Agency's good guidance practices.
Submit either electronic or written comments on the guidance at any time. General comments on Agency guidances are welcome at any time.
Submit written requests for single copies of the guidance document entitled “Class II Special Controls Guidance Document: Low Level Laser System for Aesthetic Use” to the Division of Small Manufacturers, International, and Consumer Assistance, Office of Communication, Education and Radiation Programs, Center for Devices and Radiological Health (CDRH), Food and Drug Administration, 10903 New Hampshire Ave., Bldg. 66, rm. 4613, Silver Spring, MD 20993-0002. Send one self-addressed adhesive label to assist that office in processing your request, or fax your request to 301-847-8149. See the SUPPLEMENTARY INFORMATION section for information on electronic access to the guidance.
Submit electronic comments on the guidance to http://www.regulations.gov. Submit written comments to the Division of Dockets Management (HFA-305), Food and Drug Administration, 5630 Fishers Lane, rm. 1061, Rockville, MD 20852.Start Further Info
FOR FURTHER INFORMATION CONTACT:
Richard Felten, Center for Devices and Radiological Health, Food and Drug Administration, 10903 New Hampshire Ave., Bldg. 66, rm. 1436, Silver Spring, MD 20993-0002, 301-796-6392.End Further Info End Preamble Start Supplemental Information
Elsewhere in this issue of the Federal Register, FDA is publishing a final rule classifying low level laser systems for aesthetic use into class II (special controls) under section 513(f)(2) of the Federal Food, Drug, and Cosmetic Act (the FD&C Act) (21 U.S.C. 360c(f)(2)). This guidance document will serve as the special control for low level laser systems for aesthetic use. Section 513(f)(2) of the FD&C Act provides that any person who submits a premarket notification under section 510(k) of the FD&C Act (21 U.S.C. 360(k)) for a device that has not previously been classified may, within 30 days after receiving an order classifying the device in class III under section 513(f)(1) of the FD&C Act, request FDA to classify the device under the criteria set forth in section 513(a)(1) of the FD&C Act. FDA shall, within 60 days of receiving such a request, classify the device by written order. This classification shall be the initial classification of the device. Within 30 days after the issuance of an order classifying the device, FDA must publish a notice in the Federal Register announcing such classification. Because of the timeframes established by section 513(f)(2) of the FD&C Act, FDA has determined, under § 10.115(g)(2) (21 CFR 10.115(g)(2)), that it is not feasible to allow for public participation before issuing this guidance as a final guidance document. Thus, FDA is issuing this guidance document as a level 1 guidance document that is immediately in effect. FDA will consider any Start Printed Page 20993comments that are received in response to this notice to determine whether to amend the guidance document.
II. Significance of Guidance
This guidance is being issued consistent with FDA's good guidance practices regulation (21 CFR 10.115). The guidance represents the Agency's current thinking on low level laser systems for aesthetic use. It does not create or confer any rights for or on any person and does not operate to bind FDA or the public. An alternative approach may be used if such approach satisfies the requirements of the applicable statute and regulations.
III. Electronic Access
Persons interested in obtaining a copy of the guidance may do so by using the Internet. A search capability for all CDRH guidance documents is available at http://www.fda.gov/MedicalDevices/DeviceRegulationandGuidance/GuidanceDocuments/default.htm. Guidance documents are also available at http://www.regulations.gov. To receive “Class II Special Controls Guidance Document: Low Level Laser System for Aesthetic Use,” you may either send an e-mail request to email@example.com to receive an electronic copy of the document or send a fax request to 301-847-8149 to receive a hard copy. Please use the document number 1735 to identify the guidance you are requesting.
IV. Paperwork Reduction Act of 1995
This guidance refers to previously approved collections of information found in FDA regulations. These collections of information are subject to review by the Office of Management and Budget (OMB) under the Paperwork Reduction Act of 1995 (44 U.S.C. 3501-3520). The collections of information in 21 CFR part 807, subpart E, have been approved under OMB control number 0910-0120; the collections of information in 21 CFR part 820 have been approved under OMB control number 0910-0073; and the collections of information in 21 CFR 801 have been approved under OMB control number 0910-0485.
Interested persons may submit to the Division of Dockets Management (see ADDRESSES) either electronic or written comments regarding this document. It is only necessary to send one set of comments. It is no longer necessary to send two copies of mailed comments. Identify comments with the docket number found in brackets in the heading of this document. Received comments may be seen in the Division of Dockets Management between 9 a.m. and 4 p.m., Monday through Friday.Start Signature
Dated: April 7, 2011.
Acting Assistant Commissioner for Policy.
[FR Doc. 2011-8945 Filed 4-13-11; 8:45 am]
BILLING CODE 4160-01-P